You can download all user manuals, even of older Bosch tools. To find the correct user manual, just enter the part number of your tool (it is indicated on the. : Bosch GLR Laser Distance Measurer (Discontinued by but no, the worst problem with the glr manual is there are a couple of drawings in . Manual Bosch GLR Uploaded by Mario Garcia. Copyright: © All Rights Reserved. Download as PDF, TXT or read online from Scribd. Flag for inappropriate.
|Published (Last):||3 April 2011|
|PDF File Size:||19.64 Mb|
|ePub File Size:||20.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 21 with varying temperatures or surfaces, especially over longer indirectly received reflections can distances. As an example, the user can move from a wall to the required distance, while the actual distance can be read To print the manual completely, please, download it.
Maintenance And Service Issue Remedy Measuring result not plausible Wrong reference point set Select reference point that corresponds to measurement Obstruction in path of laser beam Laser point must be completely on target surface The rangefinder section with a length of approx.
Page 7 unintentionally blind other persons. Page 11 Automatic switch-off after approx. By pressing ‘print’ button you will gor225 only current page. DO NOT stare into the laser light source.
BOSCH GLR OPERATING/SAFETY INSTRUCTIONS MANUAL Pdf Download.
Never aim light at another person or object other than the workpiece. Enter text from picture: Area Measurement Afterwards, measure the length, width and the height, one after For area measurements, push another, in the same manner as button 5 until the indicator for for a length measurement.
Only use Fewer measurements g,r225 batteries from one brand and possible when using 1. Page 15 Area Measurement Afterwards, measure the length, width and the height, one after For area measurements, push another, in the same manner as button 5 until the indicator for for a length measurement. Correct results are measurement, the distance B is achieved only when the laser calculated automatically.
Bosch GLR225 Operating/safety Instructions Manual
The area measurement laser beam remains switched on appears in the display. Don’t have an account? Issue Remedy Measuring result not plausible Wrong reference point set Select reference point that corresponds to measurement Obstruction in path of laser beam Laser point must be completely on target surface The rangefinder section with a length of approx.
Instrucciones de funcionamiento y seguridad. Move the laser back and forth To start the measurement, briefly push the measuring button To obtain this G,r225 Refund or Replacement, your return must be accompanied by the original receipt for purchase of the laser or measuring tool product.
These In combination with the laser effects are due to physical beam, the vial level 6 is not reasons and can therefore not suitable for leveling. Page 19 all measuring functions.
Laser light can damage your eyes. Got it, continue to print.
Page 18 at the bottom in the display, while the rangefinder can be moved the surface is indicated at the top. Automatic switch-off after approx.
Measuring Procedure The measured value appears The rangefinder offers a variety after 0. Repairs performed liquids, gases or dust.
Page of 25 Go. Comments to this Manuals Your Name. Page 14 To select the reference point, push button Don’t show me this message again. Table of Contents Add to my manuals Add.
To select the reference point, push button Always manul all batteries at the same time. Push the memory retrieve button 2 in order to display the value stored in the memory. Page 17 target surface is available as a After completing the second reflector. If you are not completely satisfied with the performance of your laser or measuring tool product, for any reason, you can return it to BOSCH within 30 days of the date of purchase for a full refund or replacement.
Each Do not point the time after switching on, the rear laser beam at persons or edge of the rangefinder is preset animals and do not look into Use only alkali-manganese or rechargeable batteries.